Swivel Caps need to be cleaned by hand using a non abrasive, neutral pH, mild detergent that does not contain a sheeting agent.
- Remove o-ring from cap.
- Inspect gasket for signs of wear.
- Spray and wipe the o-ring with 70% IPA (isopropyl alcohol).
- Rinse with Dionized Water.
- Replace o-ring in cap.
- Autoclave to sanitize.
SafeWASTE & Swivel Cap Autoclaving Tips:
- Recommended cycle for autoclaving is 121ºC and 15psi for 20 minutes.
- Remove cap from carboy or container prior to autoclaving.
If the containers are not vented properly during the autoclaving and cooling process, a partial vacuum could form causing the bottle to collapse. The cap should remain disengaged until the carboy reaches room temperature.
Swivel Caps may appear cloudy after autoclaving due to adsorption of water. Cloudiness will dissipate as the plastic dries, but can be accelerated using a dyring oven.
|Heavy Duty Carboy:||10 Liter HDPE|
|Swivel Cap Size:||83mm|
|Total Volume (max):||10L|
Do Not store strong oxidizing agents in carboys.
Do Not place any plastic labware in a flame.
Do Not mix any chemicals that may result in a thermic reaction that can cause product failure.
Consult local fire codes prior to storage of flammable liquids in carboys.
Solvent and Vapor Containment System for HPLC. Includes one polypropylene “Swivel Cap™ Kit” (83mm screw cap with turning wings for tightening, independent center tubing holder and solvent resistant gasket), one SafeVENT™ carbon filter with one 90° elbow fitting, 6 each male nut fittings & ferrules for 1/8th tubing (blue, not shown above), 6 each male nuts & ferrules for 1/16th tubing (green) & 6 solid male nut fittings and a 10 liter heavy duty, BPA free, (HDPE) polyethylene carboy with 1 each solid 83mm polypropylene screw cap for safe transport.
Swivel Cap™ with 83mm screw threads is for use with most HD SafeWASTE™ HPLC Solvent management systems. Caps can be applied or removed from carboy without turning the tubing holder making it very convenient to change full carboys without re-plumbing cap or twisting waste tubing during removal. With ease of use, compliance by lab personnel is increased.
The cap's center tubing holder has 6, ¼-28 screw threaded ports that accommodate the supplied, low pressure male nuts to securely hold waste tubing coming from the HPLC instrument producing a closed system that will prevent exposure to harmful organic vapors. Solid plug fittings are used on the center tubing holder when some of the ports are not in use. This closed system, when used with SafeWASTE™ containers also prevents spill incidents from HPLC solvent waste.
The included SafeVENT™ is a carbon filter that traps (VOC) organic vapors and safely vents the closed system preventing pressure build up while keeping venting safe for employees. Replacement SafeVENT filters are available.
A Use Indicator for the SafeVENT™ that indicates when a replacement is needed is available and sold separately.
* Other tube holders that fit this cap for larger size tubing are sold separately.
10 Liter SafeWASTE™ System
- Brand: Chrom-Supply Company
- Product Code: CM10-332-3411-1